Wiring the Line Power
Terminal block TB3 contains connections for the line power to the
CGA 351. To wire the line power, complete the following steps:
Before proceeding with the section, verify that the line
power has been turned off at the external disconnect
1. Attach a cable or conduit with the three line power conductors to
an entry port on the bottom of the electronics enclosure.
Be sure that the input voltage to the CGA 351 complies
with the value specified at the time of purchase. If the
voltage is incorrect, contact the factory immediately.
2. Connect the line power leads to terminal block TB3 as follows:
a. Connect the line power lead (black) to pin L.
b. Connect the neutral power lead (white) to pin N.
c. Connect the ground power lead (green) to pin G.
This completes the wiring of the CGA 351 analyzer. Proceed to
Chapter 3, Operation, for instructions on using the meter.